PAGE 63 — INTERAPID CONTACT POINT INSTALLATION
PROFESSIONAL REPAIRS & SALES IN THE U.S.A. SINCE 1959
We are proud to be a Brown & Sharpe and TESA worldwide certified service partner
Here we are comparing it to a new point since it looks like the customer may have installed a generic point. The length matches, and all is well.
Here you can see a big mistake. The user has screwed the contact point into the rounded side of the pivot. If we do this, we will no longer have an accurate indicator because the contact point will be too long.
Check that the contact point is tightly screwed into the flat side of the pivot. Although contact point wrenches are included with Interapid test indicators, we find that our small pliers work just as well.
Note: you do not need to disassemble the indicator to install a new point. Our photo is only designed to illustrate.
This is wrong! Always make sure the point is screwed into the flat side of the pivot.
Disclosure: Although most of the links on this website allow you to make a purchase directly from us, some links may direct you to Amazon. Any purchases you make from those links will result in a commission for us. That commission helps to pay for the construction and upkeep of this web site. We are grateful for any purchases you make through those links.