"American Made" light weight test indicators by one of America's original gage manufacturers. They're about as American as hot dogs and apple pie.
Do not confuse these with the lower-cost Starrett models (3808A for example) which are made in China. The Chinese versions are decidedly inferior in every way. If there is no country of origin printed on the dial, then they are made in China.
With an easy to read tilted dial similar to Swiss made Interapid indicators. The contact point angle must be held at a 15° angle to the work surface for accurate reading. This feature sets it apart from all other test indicators (except Interapid) and can prove very useful. If your shop uses other brands of indicators as well, be sure to instruct everyone to heed the 15-degree angle requirement. Plastic bezel and crystal ride on a rubber o-ring making these easy to remove and replace without special tools. They can be bought as replacement parts directly from Starrett or from Starrett distributors. Three dovetails can accept any mounting stem or holder even if they are not Starrett. Contact point length can be adjusted for calibration with a simple set screw although this is not needed on new indicators. The right half of the dial is shaded with a yellow-orange stripe for use with a mirrored set-up.
Indicators come supplied with:
Attachments (included with certain models, see chart below):
B709BCZ comes with attachments.
Easy to read tilted dial similar to Swiss made Interapid indicators. The contact point angle must be held at a 15° angle to the work surface for accurate reading.
|Range||Graduations||Point Length||Attachments||Dial Color||Model No.||Starrett No.||Price US$|
|.010"||.0001"||20 mm||none||white||708A||708AZ||check discount|
|.010"||.0001"||20 mm||none||red||708A||R708AZ||check discount|
|.010"||.0001"||20 mm||none||black||708A||B708AZ||check discount|
|.010"||.0001"||20 mm||yes||white||708A||708ACZ||check discount|
|.010"||.0001"||20 mm||yes||red||708A||R708ACZ||check discount|
|.010"||.0001"||20 mm||yes||black||708A||B708ACZ||check discount|
|.020"||.0001"||20 mm||none||white||708B||708BZ||check discount|
|.020"||.0001"||20 mm||yes||white||708B||708BCZ||check discount|
|.030"||.0005"||20 mm||none||white||709A||709AZ||check discount|
|.030"||.0005"||20 mm||yes||white||709A||709ACZ||check discount|
|.030"||.0005"||20 mm||yes||red||709A||R709ACZ||check discount|
|.030"||.0005"||20 mm||yes||black||709A||B709ACZ||check discount|
|.060"||.0005"||20 mm||none||white||709B||709BZ||check discount|
|.060"||.0005"||20 mm||yes||white||709B||709BCZ||check discount|
According to the Starrett web site, these gages "meet or exceed ISO accuracy requirements" but they do not specify or publish what those accuracies are.
The Starrett 708 and 709 series of test indicators have some idiosyncrasies that can make these repairs challenging. Unless you have experience and tools, we strongly urge you to leave this in the hands of professionals. Only the clear plastic cover (crystal and bezel assembly) and the contact points are easily replaced by the end-user (refer to notes at bottom of this page).
Spare parts are not available from us because we are not Starrett distributors. If you can not find a distributor who is willing to sell parts (there is little money to be made selling parts) then you can contact Starrett directly. Check their web site for a parts breakdown and contact information.
It is important to remember that these Starrett indicators are accurate when the contact point has an angle of about 10 to 15 degrees to the test surface. If, during calibration, you find that you are losing or gaining distance (even though the contact angle is correct) you will need to shorten or lengthen the contact point. The longer the point is, the less travel it will register. To do this, you will have to adjust the set screw which you will see on the opposite side of the pivot. Loosen the contact point first and then adjust the set screw. Now you can tighten the contact point again. Quite a bit of trial and error will get you the results you're after. If this is too much of a headache, next time don't buy Starrett.
If you need to remove the guts from the body you will have to remove adjusting screw #8 first. Similarly, if you need to remove the dial head assembly #11 you will have to remove (or at least loosen significantly) adjusting screw #36 on the cover side. You can only find this screw by removing the cover. A very small jeweler's screw driver will be required.
Q: I am hoping that you can help me out with this. One of our guys picked up a used Starrett indicator. We are trying to get the original contact for it. I ordered the MTC58-04 from your site. From what I could tell this should have been the right one but the threads are different. Would you be able to tell me what tip it needs?
There is no question that Amazon can offer you the best prices on almost every kind of precision measuring tool. They also offer easy no-questions-asked returns and often free shipping. The small distributor simply cannot compete. That's why we include this link so you can go directly to Amazon and see what they're offering.
Two things to consider:
Disclosure: Although most of the links on this website allow you to make a purchase directly from us, some links may direct you to Amazon. Any purchases you make from those links will result in a commission for us. That commission helps to pay for the construction and upkeep of this web site. We are grateful for any purchases you make through those links.
Privacy: We do not collect any information while you are visiting this website unless you proceed to place an order. If you place an order, your information will be kept confidential and will not be sold or made available to any other entity. If one of our links leads to Amazon, their privacy statement should be considered.