The dial indicator has .001" graduations with a measuring range of .4 inch. The throat depth is about 1-1/8 inch, as shown in the schematic above. Model 7328S is often in short supply. (The digital version, model 547-528, was discontinued in 2014 and there are currently no plans for a replacement.)
- Mitutoyo model 7328S dial thickness gage ... special price* $199.00 order now
- *while supply lasts - we only have one piece left at this price
Both contact points and anvils are interchangeable on Mitutoyo dial thickness gage 7328S. When you replace them, you will have to reset the indicator to zero.
Eight different contact point and anvil combinations are possible and the manufacturer guarantees accuracy for each of these. The contact point is represented by B, C and E in these diagrams. The anvil by A, D and F.
You will not be able to use a flat contact point with a flat anvil at the same time because there is no guarantee that they will be parallel to each other. If that is what you need, then you will have to get a thickness gage which has permanent, flat anvils. For options see page 12.
All six upper contact points and lower anvils are included. If the anvils become worn, replacement parts are available from many Mitutoyo distributors.
- Flat Point Ø .20" ... 133017
- Shell Point .16" radius ... 21AAA033
- Spherical Point .36" radius ... 21BZA767
- Flat Point Ø .50" ... 101188
- Needle Point .016" radius ... 21AAA030
- Shell Point .16" radius ... 21AAA032
If you use two rounded anvils (a spherical or pointed anvil with another spherical or pointed anvil) then you won't be able to set zero in the closed position. You will have to use a gage block to set the indicator at some other point, such as .25" for example. A selection of gage blocks is available on page 164.